Product Category
 Automatic Test Equipment
 Clamp Meters
 Current Probes  Current Sensors
 Data Acquisition / Oscilloscopes / Memory Recorders
 Digital Multimeters
 Ground Testers Voltage Detectors Phase Detectors
 Insulation Testers
 LCR Meters / Impedance Analyzers
 Multichannel Data Loggers
 Power Meters / Power Analyzers
 Power Quality Analyzers / Power Loggers
 Resistance Meters / Battery Testers
 Super Megohmmeters  Electrometers  Picoammeters
 Temperature Data Loggers / Compact Loggers
Impedance Analyzers, up to 3 GHz
IMPEDANCE ANALYZER IM7587• Z, L, C, R testing • Testing source frequency: 1 MHz to 3 GHz • Measuring time: 0.5 ms • Measure LCR and conduct frequency sweeps simultaneously


IMPEDANCE ANALYZER IM7580A1MHz to 300MHz Measurement Frequency Impedance Analyzer with 0.5ms Test Speed and Superior Repeatability • Z, L, C, R testing • Testing source frequency: 1 MHz to 300 MHz • Measuring time: 0.5 ms • Measure LCR and conduct frequency sweeps simultaneously


IMPEDANCE ANALYZER IM7581• Z, L, C, R testing • Testing source frequency: 100 kHz to 300 MHz • Measuring time: 0.5 ms • Measure LCR and conduct frequency sweeps simultaneously


IMPEDANCE ANALYZER IM7585• Z, L, C, R testing • Testing source frequency: 1 MHz to 1.3 GHz • Measuring time: 0.5 ms • Measure LCR and conduct frequency sweeps simultaneously


IMPEDANCE ANALYZER IM7583• Z, L, C, R testing • Testing source frequency: 1 MHz to 600 MHz • Measuring time: 0.5 ms • Measure LCR and conduct frequency sweeps simultaneously


CHEMICAL IMPEDANCE ANALYZER IM3590• Z, L, C, R, σ, ε testing • Battery measurement • 1 mHz to 200 kHz • 2 ms measurement time


EQUIVALENT CIRCUIT ANALYSIS FIRMWARE IM9000Simple Circuit Analysis & Detailed Acceptance/Rejection DecisionMaking • Optional software for Model IM3570 • 5 equivalent circuit models • Plot frequency characteristics graph from analysis results • ColeCole plot, admittance circle display


IMPEDANCE ANALYZER IM3570• Z, L, C, R testing • 4 Hz to 5 MHz • 0.5 ms measurement time • Measure LCR and conduct frequency sweeps simultaneously

LCR Meters, up to 8 MHz
LCR METER IM3536• Z, L, C, R testing • DC, or 4 Hz to 8 MHz • Accuracy guaranteed range from 1mΩ


LCR METER IM3523• Z, L, C, R testing • Testing source frequency: 40 Hz to 200 kHz • Measuring time: 2 ms


LCR METER IM3533• Z, L, C, R testing • Testing source frequency: 1 mHz to 200 kHz • Transformer measurement mode


LCR HiTESTER 351150• Z, L, C, R testing • Testing source frequency: 120 Hz or 1 kHz • Measuring time: 5 ms

Capacitance Meters, up to 1 MHz
C METER 350610• C, D, Q, low capacitance testing • 1 kHz, 1 MHz • 1.5 ms measurement time at 1 MHz


C HiTESTER 3504• C, D, large capacitance MLCC testing • 120Hz or 1kHz • 2 ms measurement time

Test fixture or Probes
SMD TEST FIXTURE IM9201• Direct connection twoterminal measurement type • DC to 3 GHz • Measurable sample sizes: 0201 to 1210 (EIA)


SMD TEST FIXTURE IM9110• Direct connection twoterminal measurement type • DC to 1 MHz • Measurable sample sizes: 008004 (EIA)


SMD TEST FIXTURE IM9100• Direct connection type • SMDs with electrodes on the bottom • DC to 8MHz • Measurable sample sizes: 01005 to 0402 (EIA)


PINCHER PROBE L2001• DC〜8MHz • 50Ω • Tip electrode spacing: 0.3 (0.01 in) to 6 mm (0.24 in)

This section provides basic knowledge about LCR meters and Impedance Analyzers measurement principles and methods of use. In addition, typical components measured with LCR meters are listed below. Click on your component to learn more about how to use LCR meters and impedance analyzers to measure it effectively and accurately.
01. LCR meter measurement principles

LCR meter basic measurement principles

LCR meters are measuring instruments that measure a physical property known as impedance. Impedance, which is expressed using the quantifier Z, indicates resistance to the flow of an AC current. It can be calculated from the current I flowing to the measurement target and the voltage V across the target’s terminals. Since impedance is expressed as a vector on a complex plane, LCR meters measure not only the ratio of current and voltage RMS values, but also the phase difference between current and voltage waveforms.

LCR meter measurement circuit: Automatic balance bridge method

The automatic balance bridge method is a circuit design that is used in many LCR meters as the measurement circuit. The circuit has four terminals (Hc, Hp, Lp, and Lc), all of which are connected to the measurement target. See the right for an overview of the circuit and below for a description of the functionality of each terminal.
Hc:
Applies a measurement signal generated with controlled frequency and amplitude to the measurement target. The frequency can be controlled within the range of several millihertz to several megahertz, and the amplitude from 5 mV to 5 V.Hp:
Detects the measurement target’s Hi potential. The detection circuit’s input impedance is extremely high, allowing accurate detection of the potential without a voltage drop.Lp:
Detects the measurement target’s Lo potential.Lc:
Converts the current flowing to the measurement target into a voltage based on the detected resistance and detects the result. The Lc terminal’s potential is always held to 0 V. 
LCR meter measurement circuit: Twoterminal method, fiveterminal method, and fourterminalpair method
02. Using an LCR meter: Basic knowledge

Typical equations for LCR meters

Equivalent circuit mode

Open correction and short correction

The test fixture used when measuring a target has residual components and can be expressed using an equivalent circuit such as that shown in the figure below. Consequently, the measured value Zm is expressed using an equation that contains these residual components, as shown below. To calculate the true value Zx, it is necessary to calculate the open residual component and short residual component and then correct the measured value. These correction processes are known as open correction and short correction, respectively, and LCR meters include functionality for performing both.
Zm:
Measured valueZs:
Short residual impedance (Rs: residual resistance; Ls: residual inductance)Yo:
Open residual admittance (Go: residual conductance; Co: stray capacitance)Zx:
True value (measurement target’s impedance) 
Measurement signal level

The measurement signal output from the LCR meter is voltagedivided between the output impedance R and the measurement target Zx. Thus the set measurement signal level V is not applied asis to the measurement target Zx. LCR meters have three measurement signal modes.
Openvoltage (V) mode:
The user sets the measurement signal level V in the figure. This value is the voltage when the measurement terminals are in the open state.Constantvoltage (CV) mode:
The user sets the value Vx in the figure (the voltage between the measurement target Zx’s terminals). This mode is used when measuring targets that exhibit voltage dependence, for example MLCCs with a high dielectric constant.Constantcurrent (CC) mode:
The user sets the value I in the figure (the current that flows to the measurement target Zx). This mode is used when measuring measurement targets that exhibit current dependence, for example inductors with cores.
03. Multilayer Ceramic Capacitors (MLCC)

What are MultiLayer Ceramic Capacitors (MLCC)?

There are two types of MLCC: a highdielectricconstant type whose capacitance varies with the measurement voltage and a temperaturecompensated type whose capacitance does not vary. The measurement conditions used when defining capacitance are set forth by separate JIS standards for temperaturecompensated and highdielectricconstant MLCCs.

Setting example of measurement conditions

Parameters Large capacitance:CsD, small capacitance:CpD Frequency See the table below DC bias OFF Signal level Rated voltage or less Measurement range AUTO Speed SLOW2 LowZ mode OFF *Otherwise, default settings are used. *The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.
IEC 6038421 Fixed surface mount multilayer capacitors of ceramic dielectric(JIS C510121) Class 1: Temperature compensating type (EIA type C0G, JIS type CH etc.)(IEC3038421)
Parameters Rated capacitance Rated voltage Measurement frequency Voltage*1 DC bias *2 C,D（tanδ） C≦1000pF All 1MHz or 100kHz
(Reference 1MHz)5Vrms or less _ C＞1000pF 1kHz or 100kHz
(Reference 1kHz)IEC 6038422 Fixed surface mount multilayer capacitors of ceramic dielectric(JIS C510122) Class 2: High dielectric constant type (EIA type X5R, X7R, JIS type B, F etc.)(IEC3038422)
Parameters Rated capacitance Rated voltage Measurement frequency Voltage*1 DC bias *2 C,D（tanδ） C≦100pF All 1MHz 1.0±0.2Vrms _ 100pF＜C
≦10μF6.3V or more 1kHz 1.0±0.2Vrms 6.3V or less 1kHz 0.5±0.2Vrms C＞10μF All 100Hz or 120Hz 0.5±0.2Vrms *1 The measurement voltage (i.e., the voltage applied to the sample) is the voltage obtained by dividing the openterminal voltage by the output resistance and the sample.
*1 The measurement voltage (i.e., the voltage applied to the sample) can be calculated based on the openterminal voltage, the output resistance, and the sample’s impedance.
*2 CV mode is convenient when measuring a sample whose impedance is unknown and when measuring multiple samples that exhibit a large degree of variability.

Highdielectricconstant capacitors

Capacitors bearing temperature characteristics such as B, X5R, and X7R use highdielectricconstant materials.
While highdielectricconstant capacitors can deliver high capacitance in a small package, their capacitance tends to vary greatly with the measurement voltage and temperature.

Instruments for Mass Production Applications

Instruments for Research &amp; Development Applications

Model Measurement frequency Features IM3570 DC,4Hz to 5MHz Frequency sweep with analyzer mode For more information, please see the individual product catalogs.

Selecting Parameter, Cs or Cp

Impedance according to frequency (when D is sufficiently small): please see table on right.
Equivalent circuit of capacitors: Large capacitance capacitors: Rp can be ignored since impedance of C is low. Select series equivalent circuit modes. Small capacitance capacitors: Rs can be ignored since impedance of C is high. Select series equivalent circuit modes. Generally speaking, series equivalent circuit mode is used when measuring lowimpedance elements (approximately 100Ω or less) such as highcapacity capacitors, and parallel equivalent circuit mode is used when measuring highimpedance elements (approximately 10 kΩ or greater) such as lowcapacity capacitors. An actual capacitor will behave as though Rs and Rp have been connected in series and in parallel, respectively, with the ideal capacitor C, as in the figure. Rp is usually extremely large (megaohmorder or greater), and Rs is extremely small (several ohms or less). An ideal capacitor’s reactance can be calculated using the following equation based on its capacitance and frequency: Xc＝1/j 2πf C[Ω]. When Xc is small, the impedance when Rp is placed in parallel can be considered to be approximately equal to Xc. On the other hand, because Rs cannot be ignored when Xc is small, the overall setup can be treated as a series equivalent circuit with Xc and Rs. By contrast, when Xc is large, Rp cannot be ignored but Rs can, so the setup can be treated as a parallel equivalent circuit. 
OpenCircuit Voltage Mode (V) and Constant Voltage Mode (CV)

The noload voltage is the voltage at the Hc terminal when no sample is connected. The voltage applied to the sample is the result of dividing the noload voltage by the output resistance and the sample.
In constantvoltage (CV) mode, the operator sets the voltage across the sample. The IM35xx reads the voltage monitor value and generates a CV by applying feedback in software. Since the 3504xx generates a CV in hardware (using an analog circuit), that instrument is capable of constantvoltage measurement at high speeds. Although the 350610 offers only noload voltage (V) mode, it has lower impedance than other models for samples for which the openterminal voltage is approximately equal to the measurement voltage due to its low output resistance (1Ω for 2.2 mF and greater ranges at 1 kHz and 20Ω for other conditions).
*1 The output impedance varies depending on the model and on whether lowimpedance highprecision mode has been enabled. Please refer to the product specifications in the instruction manual.
04. Electrolytic capacitors

How are electrolytic capacitors measured?

The measurement conditions used to define an electrolytic capacitor’s capacitance are set forth in IEC standards, and the nominal values cited by capacitor manufacturers are measured values obtained in accordance with those standards. However, because the capacitance values of electrolytic capacitors vary greatly with the measurement frequency, capacitance values should be checked at the frequency at which the circuit in question will actually be used.
Measure the equivalent series resistance (ESR), which includes factors such as the resistance of the electrolytic capacitor’s internal electrodes and the electrolyte resistance, and the tangent D (tanδ) of the loss angle under the same conditions as the capacitance.

Setting example of measurement conditions

Parameters CsDRs Frequency 120Hz, frequency at which circuit will actually be used DC bias ON 1.0V Signal level 0.5Vrms Measurement range AUTO Speed SLOW2 LowZ mode ON *Otherwise, default settings are used.
*The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.Parameters Rated capacitance Rated voltage Measurement frequency Measurement voltage*1 DC bias *2 C,D（tanδ）
Rs(ESR)All All 100Hz or 120Hz 0.5Vrms 0.7 to 1.0V *1 The measurement voltage (i.e., the voltage applied to the sample) is the voltage obtained by dividing the openterminal voltage by the output resistance and the sample.
*1 The measurement voltage (i.e., the voltage applied to the sample) can be calculated based on the openterminal voltage, the output resistance, and the sample’s impedance.
*2 DC bias need not be applied.

Low impedance high accuracy mode settings

In low impedance high accuracy mode, the instrument’s output resistance is reduced, and the measurement current is applied repeatedly for increased measurement precision. When measuring a capacitor with a high capacitance of greater than 100μF (and therefore low impedance), lowimpedance highprecision mode yields more stable measurement. The graph below compares repeatability when using the IM3570 to make measurements with lowimpedance highprecision mode enabled and disabled (100kHz, 1Ω range, 1V).
*The conditions under which lowimpedance highprecision mode can be enabled vary with the instrument model. Please refer to the instruction manual of the instrument you are using. 
Instruments for Mass Production Applications

Instruments for Research &amp; Development Applications

Model Measurement frequency Features IM3570
IM9000DC, 4Hz to 5MHz Frequency sweep with analyzer mode Optional equivalent cuircuit analysis firmware for the IM3570 IM3590 DC, 1mHz to 200kHz Can measure ESR and ESL separately with its equivalent circuit analysis function. *For more information, please see the individual product catalogs.

Equivalent series resistance (ESR) and loss coefficient D (tanδ): Circuit

The figure on the right illustrates a standard equivalent circuit for an electrolytic capacitor.
At low frequencies (50 Hz to 1 kHz), the reactance (XL) resulting from the equivalent series inductance L is extremely small and can be considered to be zero.
C: Capacitance r: Equivalent series resistance of anodic oxidation coatings R: Equivalent series resistance (ESR) L: Equivalent series inductance
*General description of Aluminum Electrolytic Capacitors (NICHICON CORPORATION)

Equivalent series resistance (ESR) and loss coefficient D (tanδ): Vector relationship

The resistance and reactance components of each element at this time are characterized by the vector relationship shown in the figure on a complex plane.
An ideal capacitor would have R = 0 and a loss coefficient D = 0, but since actual capacitors have various resistance components, including electrode foil resistance, electrolyte resistance, and contact resistance of leads and other parts, the equivalent series resistance ESR and loss coefficient D (tanδ) serve as useful indicators for use in evaluating electrolytic capacitor quality. 
Equivalent series resistance (ESR) and loss coefficient D (tanδ): Evaluation

>Since the IM3533 and IM3536 can simultaneously measure and display four parameters, they can be used to simultaneously check the reactance X, capacitance C, equivalent series resistance Rs, and loss coefficient D as indicators for use in evaluating electrolytic capacitors, as shown in the example screenshot on the right.

DC bias measurement function

Electrolytic capacitors generally are available in polarized and bipolar variants. A DC bias voltage must be applied to polarized capacitors as necessary to prevent application of a reverse voltage.
Since the IM3533 and IM3536 provide a builtin DC bias voltage function, they can apply a DC bias to capacitors, eliminating the need for an external DC power supply. 
Determining Cs and Cp

Generally speaking, series equivalent circuit mode is used when measuring lowimpedance elements (approximately 100Ω or less) such as highcapacitance capacitors, and parallel equivalent circuit mode is used when measuring highimpedance elements (approximately 10 kΩ or greater) such as lowcapacitance capacitors. When the appropriate equivalent circuit mode is unclear, for example when measuring a sample with an impedance from approximately 100Ω to 10 kΩ, check with the component’s manufacturer.
05. Tantalum capacitors

What are tantalum capacitors?

Tantalum capacitors are a type of electrolytic capacitor that uses the metal tantalum for the anode. They provide higher capacitance in a smaller package than other types of capacitors, and they offer better voltage and temperature characteristics than highcapacitance ceramic capacitors.

Setting example of measurement conditions

Parameters CsD (120Hz), Rs(100kHz) Frequency 120Hz, 100kHz DC bias OFF Signal level 0.5Vrms Measurement range AUTO Speed SLOW2 LowZ mode ON * Otherwise, default settings are used.
* The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.Surface mount fixed tantalum electrolytic capacitors with manganese dioxide solid electrolyte (IEC 603843) (JIS C51013)
Parameters Rated capacitance Rated voltage Measurement frequency Measurement voltage*1 DC bias *2 C,D(tanδ) All All 100Hz or 120Hz 0.5Vrms
or less0.7V to 1.0V Rs(ESR), Z All All 100kHz 0.5Vrms
or less0.7V to 1.0V Fixed tantalum capacitors with nonsolid electrolyte and foil electrode(IEC 6038415)(JIS C510115)
Parameters Rated voltage
Rated capacitanceMeasurement frequency Measurement voltage*1 DC bias *2 C,D(tanδ) All 100Hz or 120Hz 0.1Vp to 1.0Vp 2.1V to 2.5V *3 Rs(ESR)
ZAll Choose the frequency that yields the lowest impedance value from the following: 100 Hz, 120 Hz, 1 kHz, 10 kHz, 100 kHz, 1 MHz. 0.1Vp to
1.0Vp2.1V to
2.5V *4Surface mount fixed tantalum electrolytic capacitors with conductive polymer solid electrolyte(IEC 6038424)(JIS C510124)
Parameters Rated capacitance Rated voltage Measurement frequency Measurement voltage*1 DC bias *2 C,D(tanδ) All 2.5V
or less100Hz
or 120Hz0.5Vrms
or less1.1V to 1.5V 2.5V or greater 1.5V to 2.0V Rs(ESR),Z All All 100kHz 0.5Vrms
or lessOFF *1 The measurement voltage (i.e., the voltage applied to the sample) is the voltage obtained by dividing the openterminal voltage by the output resistance and the sample.
*1 The measurement voltage (i.e., the voltage applied to the sample) can be calculated based on the openterminal voltage, the output resistance, and the sample’s impedance.
*2 DC bias need not be applied
*3 DC bias need not be applied to bipolar capacitors.
*4 Apply only when using a measurement voltage of 0.5 Vp or greater.

Determining Cs and Cp

Generally speaking, series equivalent circuit mode is used when measuring lowimpedance elements (approximately 100Ω or less) such as highcapacitance capacitors, and parallel equivalent circuit mode is used when measuring highimpedance elements (approximately 10 kΩ or greater) such as lowcapacitance capacitors. When the appropriate equivalent circuit mode is unclear, for example when measuring a sample with an impedance from approximately 100Ω to 10 kΩ, check with the component’s manufacturer.

Instruments for Mass Production Applications

Instruments for Research &amp; Development Applications

Model Measurement frequency Features IM3570
IM9000DC, 4Hz
to 5MHzFrequency sweep with analyzer mode Optional equivalent cuircuit analysis firmware for the IM3570 IM3590 DC, 1mHz
to 200kHzCan measure ESR and ESL separately with its equivalent circuit analysis function. *For more information, please see the product catalog.

Fourterminal method

When shielding is connected close to the sample Zx, the measurement current I will return via the shielding. Because the magnetic flux generated by the current returning through the shielding negates the magnetic flux generated by the measurement current I, this technique is especially useful as a way to reduce measurement error during lowimpedance measurement (Models IM35xx).

Continuous measurement mode

The IM35xx series’ continuous measurement mode can be used to make continuous measurements while varying settings (frequency and level). In the example on the right, continuous CsD (120 Hz) and ESR (100 kHz) measurements are performed.
06. Conductive polymer capacitors

What are conductive polymer capacitors?

Conductive polymer capacitors have lower ESR (see below) than aluminum electrolytic capacitors and are characterized by greater stability with regard to temperature variations. In addition, they offer excellent stability of capacitance relative to DC bias. Measurement conditions are defined by IEC standards 60384251 and include measurements of equivalent series resistance (ESR) and the tangent D (tanδ) of the loss angle.

Setting example of measurement conditions

Parameters CsD (120Hz), Rs (100kHz) Frequency 120Hz, 100kHz DC bias ON 1.5V Signal level 0.5Vrms Measurement range AUTO Speed SLOW2 LowZ mode ON * Otherwise, default settings are used.
* The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.
IEC 60384251 Surface mount fixed aluminium electrolytic capacitors with conductive polymer solid electrolyteParameters Rated capacitance Rated voltage Measurement frequency Measurement voltage*1 DC bias *2 C,D（tanδ） All 2.5V or less 120Hz 0.5Vrms
or less1.1 to 1.5V 2.5V or more 1.5 to 2.0V Rs(ESR) All All 100kHz±10kHz 0.5Vrms
or lessOFF *1 The measurement voltage (i.e., the voltage applied to the sample) is the voltage obtained by dividing the openterminal voltage by the output resistance and the sample.
*1 The measurement voltage (i.e., the voltage applied to the sample) can be calculated based on the openterminal voltage, the output resistance, and the sample’s impedance.
*2 DC bias need not be applied.

Low impedance high accuracy mode settings

Low impedance high accuracy mode
In low impedance high accuracy mode, the instrument’s output resistance is reduced, and the measurement current is applied repeatedly for increased measurement precision. When measuring a capacitor with a high capacitance of greater than 100μF (and therefore low impedance), lowimpedance highprecision mode yields more stable measurement. The graph on the right compares repeatability when using the IM3570 to make measurements with lowimpedance highprecision mode enabled and disabled (100kHz, 1Ω range, 1V).
*The conditions under which lowimpedance highprecision mode can be enabled vary with the instrument model. Please refer to the user’s manual of the instrument you are using. 
Instruments for Mass Production Applications

Instruments for Research &amp; Development Applications

Equivalent circuit analysis function

Continuous measurement mode

The IM35xx series’ continuous measurement mode can be used to make continuous measurements while varying settings (frequency and level). In the example on the right, continuous CsD (120 Hz) and ESR (100 kHz) measurements are performed.
07. Inductors (Coils)

What are inductors or coils?

Coils may be coreless (having an air core or a core made of a nonmagnetic metal), or they may have a core made of a magnetic metal (i.e., a metal with high magnetic permeability) such as ferrite. Inductors with cores exhibit current dependence.

Setting example of measurement conditions

The phenomenon of LC resonance with the coil’s (inductor’s) inductance and parasitic capacitance is known as selfresonance. The frequency at which selfresonance occurs is known as the selfresonant frequency. When evaluating coils, be sure to measure L and Q at a frequency that is sufficiently lower than the selfresonant frequency. A coil’s inductance, which increases with frequency, can be calculated using the following equation: Z=j2πfL. To measure inductance efficiently while varying the frequency, set the measurement range to AUTO. To measure with a higher degree of precision, set the frequency to produce an impedance that can be measured with a highaccuracy range.

Setting the measurement frequency

The phenomenon of LC resonance with the coil’s (inductor’s) inductance and parasitic capacitance is known as selfresonance. The frequency at which selfresonance occurs is known as the selfresonant frequency. When evaluating coils, be sure to measure L and Q at a frequency that is sufficiently lower than the selfresonant frequency.
A coil’s inductance, which increases with frequency, can be calculated using the following equation: Z=j2πfL. To measure inductance efficiently while varying the frequency, set the measurement range to AUTO. To measure with a higher degree of precision, set the frequency to produce an impedance that can be measured with a highaccuracy range.

Setting the measurement signal level

The measurement current can be calculated from the openterminal voltage, the instrument’s output impedance, and the measurement target’s impedance. Set the measurement voltage so that the rated current is not exceeded.
When measuring a coil that exhibits current dependence (i.e., a coil with a magnetic core), set the instrument to a signal level such that the magnetic core is not saturated. When measuring a coil that does not exhibit current dependence, it is recommended to set the instrument to the signal level with the best accuracy. With the IM35xx series, the best accuracy is achieved with the V mode’s 1 V setting. With the IM758x series, the measurement signal level is defined for the power when using the DUT port’s 50 Ω termination, and the setting with the best accuracy is +1 dBm.
When measuring a coil with a core or a coil with a low rated current, the IM35xx series’ CC (constant current) mode is convenient. The measurement current is controlled in software so that it remains constant.

Instruments for Mass Production Applications

Model Measurement frequency Features IM3533 DC,40Hz to 200kHz Temperature correction function of Rdc IM3536 DC,4Hz to 8MHz Standard model,highspeed,highly stable, costeffective analyzer IM7581 100kHz to 300MHz Highspeed measurement of coils for high frequency *For more information, please see the product catalog.

Instruments for Research &amp; Development Applications

Model Measurement frequency Features IM3570 DC,4Hz to 5MHz Frequency sweep with analyzer mode *For more information, please see the product catalog.

Selecting Parameter, Ls or Lp

The table on the right shows impedance according to frequency (when D is sufficiently small)
Generally speaking, series equivalent circuit mode is used when measuring lowimpedance elements (approximately 100Ω or less), and parallel equivalent circuit mode is used when measuring highimpedance elements (approximately 10 kΩ or greater). When the appropriate equivalent circuit mode is unclear, for example when measuring a sample with an impedance from approximately 100Ω to 10 kΩ, check with the component’s manufacturer.
An inductor will behave as though the winding’s copper loss Rs and the core loss Rp have been connected to an ideal inductor L. An ideal coil’s inductance can be calculated as follows: XL＝j2πfL. Although no general formulation is possible since it varies with the magnitude of Rs and Rp, lowinductance coils are characterized by a small XL, allowing the impedance when Rp and L are placed in parallel to be treated as roughly equivalent to XL. Rs can be ignored since Ls is small, so the series equivalent circuit is used. By contrast, when the impedance is high, Rp cannot be ignored but Rs can, so the setup can be treated as a parallel equivalent circuit.
Diagram: Equivalent circuit of inductors
*Lowinductance coils: Rp can be ignored since impedance is low. Select series equivalent circuit modes.
*Highinductance coils: Rs can be ignored since impedance is high.Select series equivalent circuit modes. 
Current flowing to the coil

The current flowing to the coil can be calculated based on the openterminal voltage, the instrument’s output impedance, and the measurement target’s impedance.
*1 The output impedance varies depending on the model and on whether lowimpedance highprecision mode has been enabled. Please refer to the product specifications in the instruction manual.

Measuring Rdc

In coil evaluation, L, Q, and Rdc are measured. Instruments such as the IM3533 and IM3536 can measure L, Q, and Rdc without the need to use any other devices. After measuring L and Q with an AC signal, measure Rdc with a DC signal. *Rs and Rp are not equal to Rdc. Rs and Rp are resistance values that are measured with an AC signal. They include components such as coil loss and winding resistance, which increases due to conductor skin effects and proximity effects. When the winding material has a large temperature coefficient, Rdc will vary with temperature. The IM3533 has temperature correction functionality for Rdc.

DC superposition characteristics

Coil characteristics include DC superposition characteristics, which indicate the extent to which inductance decreases relative to DC current, an important evaluation item for coils that will be used in circuits such as power supply circuits that handle large currents.
The DC bias voltage application function built into Hioki LCR meters is designed for use in measuring capacitors, and it cannot be used to apply a DC current. To superpose a DC signal, either use the DC Bias Current Unit 9269 (or 926910) and an external power supply, or create your own circuit for the purpose.

Setting the delay time

To reduce measurement error during Rdc measurement, Hioki LCR meters cycle the generated voltage on and off to cancel the internal offset (DC adjustment function).
When the voltage being applied to the inductor changes, the output resistance and inductor’s equivalent series resistance and inductance cause transient phenomena. Set a sufficiently long delay time during Rdc measurement to ensure that the measurement results are not affected by these phenomena. The name given to the delay time setting varies by model, as does measurement timing. For more information, please see the instruction manual for the model you intend to use.

Setting the delay time  reducing variability

If you are unsure of the appropriate delay time, first set as long a delay time as possible. Then gradually shorten the delay time while verifying that measured values do not exhibit any variability.
08. Electric Transformers

What are electric transformers?

AC voltages can be stepped up or down using a transformer. In terms of their basic structure, transformers consist of primary and secondary windings around an iron core.
When current flows, a magnetic field is generated inside the windings, creating a voltage. The size of this voltage is proportional to the number of turns. For example, a primary winding (on the input side of the transformer) with 100 turns and a secondary winding (on the output side of the transformer) with 200 turns would step up an input voltage of 100 V to an output voltage of 200 V since the number of output turns is twice the number of input turns. Note that there is no change in power between the primary and secondary sides of the transformer.

Setting example of measurement conditions

Parameters Ls,Q,Rdc Frequency Selfresonant frequency or less *1 DC bias OFF(ON is NOT applicable) Signal level Rated current or less *1 Measurement range AUTO Speed SLOW2 LowZ mode OFF ※1 Cf. Inductors Application note *Otherwise, default settings are used.
*The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.
Illustration:
R1: Primary winding
R2: Secondary winding resistance
C1: Primary winding floating capacity
C2: Secondary winding floating capacityParameters for each electric transformer
The transformer is an application of an inductor, and measurement methods are the same as for other inductors. Transformer measurement includes the following principal evaluation parameters:
*Primary inductance (L1)and secondary inductance (L2)
*Leakage inductance
*Capacitance between windings (C)
*Mutual inductance (M)
*Turn ratio 
Instruments for Mass Production Applications

Instruments for Research &amp; Development Applications

Model Frequency Features IM3570 DC,4Hz to 5MHz Frequency sweep with analyzer mode *For more information, please see the product catalog.

Primary inductance (L1) and secondary inductance (L2)

As shown in the figure to the right, a measuring instrument can be connected directly to the primary or secondary side of the transformer to measure the primary or secondary inductor. However, all other windings must be left in the open state. Exercise care as inductance measurement results include the effects of the winding’s distributed capacitance.

Leakage inductance

In an ideal transformer, shorting output causes input to be shorted as well. However, in an actual transformer, leakage inductance remains even when output is shorted. As shown in the above figure, the leakage inductance can be determined by shorting the secondary side of the transformer and measuring the primary side’s inductance.

What is leakage inductance?

The magnetic flux that links the transformer’s primary and secondary windings is known as the main magnetic flux (φ12 or φ21). Apart from the main magnetic flux, the transformer’s magnetic flux also includes primary leakage flux (φs1) , which links the primary winding but not the secondary winding, and secondary leakage flux (φs2), which links the secondary winding but not the primary winding.
Although only the main magnetic flux exists in an ideal transformer, actual transformers always have magnetic leakage, and therefore leakage flux. Since this leakage flux does not link only the primary and secondary windings, it does not contribute to the transformer’s voltagemodifying operation. At the same time, the fact that the leakage flux does not link only the primary and secondary windings also means that it contributes as each winding’s inductance. In this way, the primary leakage flux acts as the primary leakage inductance, and the secondary leakage flux acts as the secondary leakage inductance.

Capacitance between windings

As shown in the figure to the right, the winding capacitance between the primary and secondary sides of the transformer can be measured by connecting each winding to the measuring instrument.

Mutual inductance

The mutual inductance can be calculated by measuring the inductance in parallel while in phase and then in series out of phase and then using the equation shown below.
M=(LaLo)/4

Turn ratio

As shown in the figure to the right, the turn ratio can be approximated by measuring the impedance value Z on the primary side of the transformer after connecting the resistance R to the secondary side.
In addition, the turn ratio can be calculated by measuring the primary inductance L1 and the secondary inductance L2. However, the value will only be an approximation due to the effects of factors such as magnetic leakage.

Transformer measurement functionality

The LCR Meter IM3533/IM353301’s transformer measurement functionality can be used to calculate the mutual inductance, turn ratio, and inductance difference. Turn ratio measurement with the IM3533/IM353301 involves measuring the primary and secondary inductance values and then calculating the turn ratio.
09. RFID (Contactless IC cards, Contactless IC tags)

How are RFID regulated?

The operating frequencies of RFIDs, which are also known as IC tags or contactless IC cards, are defined by international standards. When performing L measurement of a board used by a contactless IC card, the measurement must be made near the operating frequency of 13.56 MHz.

Setting example of measurement conditions

Measurement mode ANALYZER Parameters Zθ frequency characteristics analysis（LQ、R evaluation available） Sweep parameter FREQ Sweep frequency Sweep measurement close to the operating frequency (See the table below) Signal level V mode 1V (350x, IM35xx series) or 1dBm (IM758x series) *The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.

RFID standards

Category Frequency Effective distance Standard ID cards 13.56MHz Up to 10cm
(Proximity applications)ISO14443 Automatic recognition 125kHz Up to 70cm (Vicinity applications) ISO14443 13.56MHz ISO15693 
Structure of RFID tags

RFIDs generally consist of an antenna and IC. Signal transmission is accomplished by a resonant circuit formed by the antenna inductor (Ls) and the IC chip’s builtin input capacitance (Cp).

Frequency characteristics of defective and nondefective components

As shown in the figures, the Zθ frequency characteristics of defective and nondefective components differ. The nondefective component exhibits a resonance point near the operating frequency.

Instruments for Research &amp; Development Applications

Model Measurement frequency RFID IM758x series 100k to 1.3GHz * Mainly for highfrequency RFID IM3570 4Hz to 5MHz Mainly for lowfrequency to medium frequency RFID *For more information, please see the product catalog.

Pass/fail judgments using analyzer mode

Either of two methods can be used to generate pass/fail judgments when using analyzer mode: peak judgment and area judgment.

How to set the judgment area

Judgement areas can be set as follows.
• A knowngood element’s measured value can be used as the reference (±10% of the reference element’s measured value, etc.).
• A userspecified value can be entered (1 k±10%, etc.). 
Ascertaining electrical constants by means of equivalent circuit analysis

The instrument’s equivalent circuit analysis function can be used to calculate the constants in a threeterminal circuit model such as an RFID antenna.
*Model A should be used for coils with a large core loss (R) in order to facilitate more accurate analysis. 
Equivalent circuit models
10. Piezoelectric elements

What are piezoelectric elements?

Piezoelectric elements are used in a wide range of applications, including buzzers, sensors, and filters. Since resonant and antiresonant frequencies characterize their impedance/frequency characteristics, an impedance analyzer is the ideal instrument for use in analyzing their characteristics.

Setting example of measurement conditions

Measurement modes ANALYZER Parameters Zθ Sweep parameter FREQ Sweep frequency Set to a range within which the resonant, antiresonant frequency can be checked. Signal level Depends on the measurement items Equivalent circuit model E *The above settings apply to an example measurement. Since optimal conditions vary with the measurement target, specific settings should be determined by the instrument operator.

Equivalent circuit of piezoelectric elements

Close to its resonant frequency, a piezoelectric element can be depicted as an electrical equivalent circuit. Specifically, such an element can be depicted as a parallel capacitance CO that is connected in parallel to a series circuit consisting of the series inductance L1, the series capacitance C1, and the series resistance R1.

Actual measurement and analysis

Recommended products

Model Frequency Features IM3590 DC,1mHz to 200kHz Analyzer mode (low frequency), equivalent circuit analysis IM3570
IM9000DC,4Hz to 5MHz Frequency sweep with analyzer mode Optional equivalent circuit analysis firmware for the IM3570 IM7581 100kHz to 300MHz Analyzer mode (high frequency), equivalent circuit analysis *For more information, please see the product catalog.

Measuring resonant frequency and antiresonant frequency

The frequency fm characterized by minimum inductance and the frequency fn characterized by maximum inductance can be calculated from the element’s impedance/frequency characteristics using the instrument’s peak search function. In addition, it is possible to calculate the resonant frequency fr, which is characterized by a phase of 0, and the antiresonant frequency fa.
The series resonant frequency fs and the parallel resonant frequency fp can be expressed as shown on the right.

Equivalent circuit analysis

Pass/fail judgments using analyzer mode

Either of two methods can be used to generate pass/fail judgments when using analyzer mode: peak judgment and area judgment.
Judgement areas can be set as follows.
• A knowngood element’s measured value can be used as the reference (±10% of the reference element’s measured value, etc.).
• A userspecified value can be entered (1 k±10%, etc.).